1 APPA N/N DIGITAL MULTIMETER INSTRUCTION MANUAL 2 TABLE OF CONTENTS 1. SAFETY 1 2. PRODUCT DESCRIPTION 2 3. PANELS . APPA N, N Conversion Type: (AC + DC) True RMS reading on AIR CONDITIONING mode. ACV subsp. + % + 8 d. Input Impedance: 10 m?// pF. Centenary Materials — We offer quality service and products: APPAN APPA APPA_Series Digit Multimeters.
|Published (Last):||2 March 2012|
|PDF File Size:||1.31 Mb|
|ePub File Size:||19.63 Mb|
|Price:||Free* [*Free Regsitration Required]|
The instrument enhances More information. Didn’t find required product? To get the best service More information. Set temperature measurement default at F. Remove the battery from meter. Remove all power from the circuit before connecting the test leads.
APPA 107N/109N DIGITAL MULTIMETER INSTRUCTION MANUAL
Licensed software products are owned by Tektronix or its suppliers and are More information. Pause time is seconds, exceed pause time stores as seconds. Silicone test leads 4 mm probes. Installing improper fuses can cause zppa and product damage.
To get the best. These statements identify conditions or practices that could result in damage to the equipment or other property.
Do not operate without covers. One pair of red and black test leads One pair of 19 More information. This device can measure Earth Ground More information. Radiated and Conducted Emissions. LA 1 Contents Introduction Auto Power Off function is disabled. Remove battery cover from case bottom. Manufactured to a high standard, this product will, if used according to these instructions, More information. AEMC Instruments certifies that this instrument has been calibrated using.
Handheld Digital Multimeters Model: Licensed software products are owned by Tektronix or its suppliers and are. V3 Thank you for purchasing a Sealey product.
Read, understand and follow Safety Rules and Operating Instructions in this manual before using this product.! Suppliers will contact you. Any instrument More information. Lifetime Limited Warranty Each Fluke. To avoid damage to 17n surface of the meter, do not use abrasive or chemical cleaning agents.
Safety notes These meters have been produced to comply with category II requirements. To avoid damage to the meter, do not expose 10n to sprays, liquids, or solvents. Buzzer beeps once for every key-press. Auto-range for More information.
APPA N/N DIGITAL MULTIMETER INSTRUCTION MANUAL – PDF
Obligatory field is not filled. Specifications are subject to change without notice. Limit large current measurements 10 to 20A to 30 seconds and allow two minutes of cooling between measurements. To avoid personal injury, do not appz any voltage or current to the product without the covers in place. To avoid potential hazards, use the product only as specified.
APPA 107N Instruction Manual
Assemble the top case and PCB of meter. Follow step 1 to step 4 described in Battery Replacement.
Frequency And Duty Factor Measurements 5. Maximum, Minimum and Average indicators. RANGE function is only enable in just data log mode.
Precautions and safety information Specifications Basic maintenance. Read, understand and follow Safety Rules and Operating Instructions in this manual before using this product.